ERT cables for ABEM instruments are multi-electrode cables which are used to transfer current and potential signals from resistivity meter to the ground, which is mostly widely used in electrical resistivity tomography(ERT, Electrical Imaging) method and resistivity imaging method for infrastructure projects, environmental studies, water content variations, movements of pollutants and seepage through embankment dams etc.
Click here and download wiring diagram for a ” typical ERT cable 32 channel ” for ABEM instrument or or contact us and get more detailed information.
|Structure of Cable Conductor (Per core)||7nos.of Bare Copper Wires Stranding(Copper Wire O.D. 0.12mm)|
|Numbers of Cable Conductors||– Standard 32 conductors(16x2x7/0.12)|
– Customized cable conductors are available as per client’s requests.
|Electrode cable take-out||Molded cable take-out in the color of red or blue.|
|Cable Jacket||7.2mm OD with color of orange.|
|Min. Breaking Strength||130 kgs.|
|Insulation Resistance||≥1000 MΩ/km|
|Cross Section (Per core)||0.08mm2|
|Voltage Withstand Value (Per core)||1000V|
|Current-Carrying Capacity per core|
– ERT cables for ARES, IRIS or PASI Resistivity Meters are also available.
– Specifications are subject to change without prior notice.